Effective maintenance laboratory muffle furnace
Maintenance of all kinds of equipment, according to the actual situation we need to be processed. Need laboratory muffle furnace maintenance skills we continue to grope in the work, which is to ensure normal operation of equipment must work to be done.
1. To ensure accurate measurement, test personnel proofread every once in thermometer temperature control device with DC potentiometer, in order to reduce the error for each test.
2. The best instrument is placed in a dry and ventilated, non-corrosive gases, ambient temperature of 10-50 degrees Celsius, relative humidity less than 85%.
3. Test personnel should regularly check hotline, AC contactor contacts are loose, if necessary, immediate repair.
The silicon carbide-type furnace, silicon carbide was found damaged, replace the same specification and resistance value close to the new silicon carbide. When replacing pay attention to protect the new silicon carbide silicon carbide, allow four hours of continuous work at the highest temperature.
All maintenance work is to give equipment to provide a better working environment, but also to achieve better efficiency, while these practices also effectively extend the service life of the equipment.
Sliding tube furnace
High Temperature Vacuum furnace
High temperature high pressure tube furnace
Large Multi station tubular furnace