Temperature adjustment method of muffle furnace
The temperature adjustment method of the muffle furnace:
1. In the non-setting state, press and hold the "shift/auto-tuning" button for 6 seconds and then enter the system auto-tuning program. The setting indicator will flash. After the auto-tuning is completed, the indicator will flash intermittently, and the controller will get a set Better system PID parameters, parameter values are actively retained. The temperature of the muffle furnace
2. The system can be self-tuned when the temperature control of the muffle furnace is not ideal. During the auto-tuning process, the temperature will have a large overshoot. The user of the temperature adjustment method should fully consider this factor before performing system auto-tuning.
3. During the system auto-tuning process, long press the "shift/auto-tuning" button for 6 seconds to interrupt the auto-tuning program. The alarm indicator does not light up and the buzzer does not sound, but the heating alarm relay will be actively disconnected.
High Temperature Vacuum furnace
Sliding tube furnace
High temperature high pressure tube furnace
Large Multi station tubular furnace